Caire SeQual Eclipse 5 Portable Oxygen Concentrator
- Up to 5.4 hours of battery
- AutoSTAT technology
- CPAP compatible
Caire SeQual Eclipse 5 Portable Oxygen Concentrator Overview
The SeQual Eclipse 5 Portable Oxygen Concentrator is one of the few portable oxygen concentrators that offers both continuous flow and pulse dose oxygen settings. Continuous flow settings range from 0.5 to 3LPM (liters per minute) and can be adjusted in increments of 0.5LPM. Pulse dose settings are offered on a scale from 1- 9, with Setting 1 delivering 16mL of oxygen per pulse and Setting 9 delivering 192mL of oxygen per pulse. Both settings ensure that users can get the most out of the Eclipse 5 portable concentrator no matter what their prescription needs are.
The Caire Eclipse 5 Portable Concentrator weighs 18.4 pounds with the battery installed, making it robust enough for home use. While it is one of the heavier portable oxygen concentrators on the market, the mobile cart that is included makes it very easy to move around and allows you to continue to be mobile while using oxygen. The cart features a telescoping handle that can be easily adjusted to whatever height you need.
SeQual Eclipse 5 Portable Oxygen Concentrator - Key Features
- Automatically switches to continuous oxygen flow if breathing is not detected in pulse dose mode.
- Three varied pulse dose settings
- Pulse dose volume ranges from 16ml to 192ml
- Get 3 LPM continuous flow with prescription
- Get DC and AC power supply cords that are durable and easy to carry
Caire SeQual Eclipse 5 - AutoSAT Technology
The AutoSAT technology offered by SeQual Eclipse 5 monitors the breathing of the user and offers incremental therapy if inhalation is not identified. If the device does not sense an inspiratory effort when in pulse dose mode, a steady flow of oxygen would be administered while it tries to scan for a breath every 15 seconds. Additionally, AutoSAT technology adapts to the user’s breath rate and increases or decreases oxygen production as needed.
SeQual Eclipse 5 Portable Oxygen Concentrator - Robust Performance
The Claire SeQual Eclipse 5 has raised the performance bar for portable oxygen concentrators. It incorporates all the benefits of Eclipse 3 with next-generation internal components to make it more reliable. With its compact design, it is highly portable and the DC power operation provides the freedom to operate this portable oxygen concentrator at any available DC outlet. The Claire SeQual Eclipse 5 offers both Pulse flow settings and continuous flow settings up to 3 LPM.
This portable oxygen concentrator also provides a super setting of 192mL which can be achieved while connected to either AC or DC power. The battery can be charged even while operating at all flow settings up to 2.0 LPM with DC power.
Caire SeQual Eclipse 5 POC - High Pulse Dose Output
This portable oxygen concentrator is capable of providing the highest pulse dose oxygen output setting for even the most difficult-to-treat patients. Pulse dose settings range between 1.0 to 9.0 LPM. Pulse dose settings between 1.0 to 6.0 go up in increments of 0.5 instead of the traditional increment of 1.0 allowing for greater flexibility in treatment. Pulse dose settings between 7.0 to 9.0, also called the super-bolus settings, are adjusted with an increment of 1.0.
The unmatchable power and flexibility of the Claire SeQual Eclipse 5 provide a cost-efficient solution to costly oxygen cylinders for oxygen therapy patients. This portable oxygen concentrator can act as both a portable unit and a stationary unit and is able to meet the high demand of hospitals and clinics.
SeQual Eclipse 5 POC - Portability
Even though the Caire SeQual Eclipse 5 is a heavier oxygen concentrators on the market currently, it’s advanced features help you enjoy your oxygen therapy. A mobile cart with a telescopic handle is the greatest advantage with the Caire SeQual Eclipse 5 portable oxygen concentrator. It weighs 18.4 pounds with a cart and the battery.
Caire SeQual Eclipse 5 Portable Oxygen Concentrator - Alarm Features
In addition to the numerous benefits listed above, the Caire Eclipse 5 offers a number of alarms to let the user know when something is wrong. The alarms are classed as low-priority (yellow LED) and high-priority (red LED). Examples of some alarms include.
- External power failure
- Low oxygen flow
- System malfunction
Caire SeQual Eclipse 5 - Additional Features
If you experience any of these alarms, consult the user manual or call our customer care team for support.
For the most effective and efficient therapy, the manufacturer recommends using a nasal cannula and oxygen tubing that is no longer than 7 feet in length. Longer oxygen tubing may result in the machine’s inability to detect inspiratory effort when in pulse dose mode.
The operating volume for the Caire Eclipse 5 concentrator is approximately 48 dBA, which makes it slightly louder than a library.
The Eclipse 5 portable oxygen concentrator is FAA-approved for in-flight use. Contact your airline in advance, as each airline has a different policy.
It is FAA cleared for usage during flights, cruise ships, and other forms of transportation.
SeQual Eclipse 5 Portable Oxygen Concentrator Includes
- Sequal Eclipse 5 Portable Oxygen Concentrator
- Power Cartridge (Battery)
- AC power supply
- DC power supply
- Cart with Telescoping Handle
- 5 Nasal Cannulas
- 3 year Standard Warranty
What Is The Proper Placement Of The Caire Sequal Eclipse 5?
Select a place for the unit that prevents smoke, fumes, and contaminants from entering the device. The right positioning of the unit should allow air intake at the top rear of the cabinet via the air inlet filter and allow exhaust air to escape the exhaust vent at the bottom left of the device freely. Place the system in such a way that you can hear any alarm warnings. Place the tubing of the oxygen so that it does not kink or get tangled.
What Is The Battery Life of the SeQual Eclipse 5 Portable Oxygen Concentrator?
At most, users are able to go approximately 5 hours between charges or at minimum, approximately an hour. While it may be tempting to switch your settings to extend the life of your battery, we recommend consulting your doctor first, as you want to ensure that you are receiving the most efficient and effective oxygen therapy for your needs.
In addition to the battery, the Caire SeQual Eclipse 5 Portable Oxygen Concentrator also comes equipped with an AC power cord for use at home and a DC power cord for use on the road. If external power is interrupted for any reason, the internal battery will automatically overpower the device.
Additional batteries are available for purchase for individuals to keep on hand as spares. To recharge your battery, simply plug the SeQual Eclipse 5 into an external AC or DC power source. If your device is connected to DC power, the battery will only recharge while the concentrator is on a setting of 2.0LPM or less. The battery can take anywhere between 2 to 5 hours to charge fully. The Caire SeQual Eclipse also comes with 4 alarms that alert the user to a power malfunction, including low battery and external power failure.
How To Maintain The Caire SeQual Eclipse 5
In addition to replacing your nasal cannula and oxygen tubing on a regular basis, the Eclipse 5 oxygen concentrator requires regular maintenance of its air inlet filter. This filter requires daily inspection and cleaning once per week to ensure proper device function. To clean the filter:
- Remove the filter from the device.
- Wash it with warm water and mild detergent.
- Rinse thoroughly and squeeze out excess water.
- Allow the filter to air dry completely.
- Replace the filter in the device.
Cleaning tip: Keep a clean second filter ready as a substitute for use while cleaning and drying the first filter
The inlet filter should be replaced on an annual basis, with proper cleaning. If Eclipse 5 is being used in a particularly dusty environment, it may need more frequent cleaning and replacement.
To clean the exterior of the SeQual Eclipse 5 and its battery, make sure the device is unplugged. Use a damp cloth or sponge sprayed with mild detergent and gently wipe down the device.
Do not attempt to perform maintenance on any internal components of the device. Contact our customer care team for assistance if your device is under warranty.
How To Connect The DC Supply Connection To Caire Sequal Eclipse 5?
- The following applies when the Eclipse 5 is attached to the DC power supply in a vehicle:
- When working with the DC power supply in a vehicle, the Eclipse 5 will operate at up to 3.0LPM on continuous flow.
- The Eclipse 5 would charge the battery when in a car on a DC power supply while operating at 2 LPM or less on continuous flow.
- If the Sequal Eclipse 5 battery is charged, the icon on the battery will look like a waterfall. If the battery of the Caire Eclipse 5 is not charged, the battery symbol will look solid.
Caire SeQual Eclipse 5 Portable Oxygen Concentrator Parts
- Caire SeQual Eclipse Travel Accessory Kit$462.45Out of stock